Numerical expressions to compute g–g and g–XK coincidence summing corrections were deduced byusing a suitable computer program and a matrix representation of a decay scheme. For point sourcesonly full-energy peak and total efficiencies are needed. Alternatively, values of peak-to-total ratio can be introduced. For extended sources, the same expressions can be considered with the introduction of‘‘effective efficiencies’’. Examples of the use of the expressions for point sources and a particulate filtersample measured with a 60% relative efficiency HPGe detector are reported.
|Numero di pagine||6|
|Rivista||Applied Radiation and Isotopes|
|Stato di pubblicazione||Published - 2010|
All Science Journal Classification (ASJC) codes